 Research
 Open Access
 Published:
Sequential algorithm for life threatening cardiac pathologies detection based on mean signal strength and EMD functions
BioMedical Engineering OnLine volume 9, Article number: 43 (2010)
Abstract
Background
Ventricular tachycardia (VT) and ventricular fibrillation (VF) are the most serious cardiac arrhythmias that require quick and accurate detection to save lives. Automated external defibrillators (AEDs) have been developed to recognize these severe cardiac arrhythmias using complex algorithms inside it and determine if an electric shock should in fact be delivered to reset the cardiac rhythm and restore spontaneous circulation. Improving AED safety and efficacy by devising new algorithms which can more accurately distinguish shockable from nonshockable rhythms is a requirement of the presentday because of their uses in public places.
Method
In this paper, we propose a sequential detection algorithm to separate these severe cardiac pathologies from other arrhythmias based on the mean absolute value of the signal, certain loworder intrinsic mode functions (IMFs) of the Empirical Mode Decomposition (EMD) analysis of the signal and a heart rate determination technique. First, we propose a direct waveform quantification based approach to separate VT plus VF from other arrhythmias. The quantification of the electrocardiographic waveforms is made by calculating the mean absolute value of the signal, called the mean signal strength. Then we use the IMFs, which have higher degree of similarity with the VF in comparison to VT, to separate VF from VTVF signals. At the last stage, a simple rate determination technique is used to calculate the heart rate of VT signals and the amplitude of the VF signals is measured to separate the coarse VF from VF. After these three stages of sequential detection procedure, we recognize the two components of shockable rhythms separately.
Results
The efficacy of the proposed algorithm has been verified and compared with other existing algorithms, e.g., HILB [1], PSR [2], SPEC [3], TCI [4], Count [5], using the MITBIH Arrhythmia Database, Creighton University Ventricular Tachyarrhythmia Database and MITBIH Malignant Ventricular Arrhythmia Database. Four quality parameters (e.g., sensitivity, specificity, positive predictivity, and accuracy) were calculated to ascertain the quality of the proposed and other comparing algorithms. Comparative results have been presented on the identification of VTVF, VF and shockable rhythms (VF + VT above 180 bpm).
Conclusions
The results show significantly improved performance of the proposed EMDbased novel method as compared to other reported techniques in detecting the life threatening cardiac arrhythmias from a set of large databases.
Background
Ventricular Fibrillation (VF) and Ventricular Tachycardia (VT) are lifethreatening cardiac arrhythmias generally observed in adults with coronary artery disease. In 1979, automatic external defibrillators (AEDs) were introduced to accurately analyze the cardiac rhythms and, if appropriate, advise/deliver a highenergy shock to those patients who suffer from coarse VF and VT of a rate above 180 bpm, combinedly known as the shockable rhythms [6]. Though a significant number of works have been published on this topic, the scope for development of more accurate and reliable techniques relaxing assumptions of certain previous works and incorporating features from diverse nature of the cardiographic signals is yet open. Based on separation capability, the algorithms available in the literature can be classified into categories such as, separating VF from VT [4, 7, 8], VF from normal sinus rhythm (NSR) [9], VF plus VT from nonVTVF [10], shockable rhythms from other ECG pathologies [5, 11, 12], VF from nonVF [1–4, 13–24]. Comprehensively, the last two categories [25] are the most realistic for fruitful hospital management of cardiac abnormalities.
To separate VF from VT many efforts have been aimed at characterizing these abnormalities by means of diverse techniques such as the sequential hypothesis algorithm proposed by Thakor et al. [4], continuous wavelet transform [7], paired unipolar electrograms [8] etc. But only separating VF from VT is not useful for cardiac management. Because, in real life problems, other types of abnormalities are also present. A recent work is presented in [9] using the EMD technique to separate VF from NSR which shows almost 100% accuracy. But, when other types of pathology except the NSR and VF are present, poor accuracy is obtained. To separate VT plus VF from other arrhythmias, a time domain based complexity measure algorithm has been proposed in [10]. But it fails to show good performance due to its weakness in selecting a proper threshold value. Another approach has been reported in [5] to classify arrhythmias into two types: shockable and nonshockable signals. This work shows quite good accuracy but improvement area is still open. Various algorithms have been developed for classifying the abnormalities according to the last category. To separate VF from other arrhythmias, different methods were proposed based on different techniques of signal processing, such as the threshold crossing interval (TCI) algorithm [4], autocorrelation function (ACF) [13], probability density function method [14], VFfilter method [15], [16], [17], rate and irregularity analysis [18], [19], sequential hypothesis testing algorithm [20], [21], correlation waveform analysis [22], spectral analysis [3] and four fast template matching algorithms [23]. But these algorithms fail to show good performance when tested on a large database due to the some shortcomings in their reported algorithms. For example, the TCI method, based on a time domain technique, fails to detect the normal sinus rhythm (NSR) signal due to several factors, e.g., choice of 1s analysis window, improper threshold etc. [24]. An improved version of this algorithm called the threshold crossing sample count (TCSC) method has been reported in [24] by removing some of the drawbacks of the TCI method. But the TCSC algorithm does not consider the shape of the ECG signal, therefore, it fails to classify VT into the nonVF group. On the other hand, the ACF relies on the regularity in NSR and irregularity in VF rhythms [26]. But practically, in most cases, there is no strict regularity found in the NSR signal and, therefore, the detection accuracy of the NSR signal by this method severely falls. The spectral analysis method successfully detects the nonVF signal from ECG arrhythmias. But in the detection of VF, this method shows poor accuracy due to the false detection of the VF signal with low peak frequency in the spectrum [26]. On the other hand, the Hilbert transform (HILB) [1] and phase space reconstruction (PSR) [2] algorithms employing phase space plot of the ECG signal demonstrate improved performance of VF detection. Because the phase space plot is based on the histogram of a signal, it does not consider the shape of this signal. Thus, to separate VT from VF when other arrhythmias are also present, these two methods are not very suitable.
In this paper, we propose a sequential detection algorithm based on the mean absolute strength and certain loworder intrinsic mode functions (IMFs) of the EMD analysis of the signal along with a simple rate determination technique. In our proposed algorithm, we not only separate VF but also VT from other arrhythmias. VT plus VF (VTVF) is separated from other arrhythmias in the first stage using an index called the mean absolute value (MAV). Then we decompose the VTVF signal into IMFs using the EMD technique to discriminate VF from VT. EMD was introduced in [27] for processing signals from nonlinear and nonstationary processes. Here, we apply the EMD technique to biomedical signals and particularly for ECG analysis. Next, a simple rate determination algorithm is utilized to classify VT according to the heart rate and to separate coarse VF from fine VF, amplitude of the VF signals are measured. Finally, this sequential ECG arrhythmias classification approach is interpreted as three different detection schemes, such as, VTVF from nonVTVF; VF from nonVF; shockable from nonshockable rhythms. While proposing an algorithm for detecting the shockable rhythms special care must be taken to make the specificity high. It will then ensure the false alarm generation probability of the AEDs low. But an algorithm with high specificity generally results in low sensitivity. To mitigate this contradictory requirement, detection of the shockable rhythms using a sequential algorithm is found to be more effective. At last, in the 'Results' Section, we compare our algorithm with different wellknown algorithms available in the literature.
Methods
ECG signals
We use the MITBIH Arrhythmia Database (MITDB) [28], Creighton University Ventricular Tachyarrhythmia Database (CUDB) [29] and MITBIH Malignant Ventricular Arrhythmia Database (VFDB) [30] to evaluate our algorithm. The MITDB contains 48 files, 2 channels per file, each channel 1805 seconds long. The CUDB contains 35 files, 1 channel per file, each channel 508 seconds long. The VFDB contains 22 files, 2 channel per file, each channel 2100 seconds long. In our analysis, we choose episodes of 8s long from the whole MITBIH arrhythmia and CU databases. We perform a continuous analysis by taking the data in steps of 1 sec. Thus, the total number of 8s episodes collected from the MITDB and CUDB are (18057) × 48 × 2 = 172608 and (5087) × 35 = 17535, respectively. Since, the VFDB includes ECG recordings of subjects who have experienced episodes of sustained VT and VF, we use this database for VF and VT episodes. By taking the ECG signal in steps of 1 sec we choose 4000 episodes of VF and 4000 episodes of VT from this database. Therefore, a total of 172608 + 17535 + 4000 + 4000 = 198143 episodes are used to compare our algorithm with other algorithms. Amongst these 198143 episodes, we have noticed some episodes which are annoted as the noise signals. Since, in this work we have no interest in these noise signals, we have omitted these noise episodes. Also, analysis of the distinct mode asystole signal is not presented here. Therefore, this type of ECG signal is not included into our complete dataset.
The complete dataset includes the following types of ECG signals.

1.
Normal beat

2.
Left bundle branch block beat (LBBB)

3.
Right bundle branch block beat (RBBB)

4.
Atrial premature beat (APC)

5.
Aberrated atrial premature beat

6.
Nodal (junctional) premature beat

7.
Supraventricular premature or ectopic beat

8.
Premature ventricular contraction (PVC) beat

9.
Fusion of ventricular and normal beat

10.
Atrial escape beat

11.
Nodal (junctional) escape beat

12.
Paced beat

13.
Fusion of paced and normal beat

14.
Unclassifiable beat

15.
Blocked APC

16.
Ventricular tachycardia

17.
Ventricular fibrillation
To determine the discriminating threshold and verify its effectiveness, the complete dataset is divided into two subsets: training and test datasets. The training dataset is used to determine the threshold value. To check the efficacy of the threshold value determined from the training dataset, the test dataset is used. Both the datasets include all types of above mentioned rhythms. The training dataset includes:

1.
(1805  7) × 23 × 2 = 82708 episodes from MITDB (file no. 100109, 111119, 121124).

2.
2000 episodes of VF and 2000 episodes of VT from VFDB.
On the other hand, the test dataset includes:

1.
(1805  7) × 25 × 2 = 89900 episodes from MITDB (file no. 200203, 205, 207210, 212215, 217, 219223, 228, 230234).

2.
2000 episodes of VF and 2000 episodes of VT from VFDB.
Classification of the ECG signals according to the AHA recommendations
According to the AHA recommendations, all ECG abnormalities are classified into following categories [6]:

1.
Shockable rhythms

'coarse VF': any VF signal with an amplitude of > 200 μV.

'VThi': rapid ventricular tachycardia with a rate of > 180 bpm.

2.
Nonshockable rhythms

'NSR': normal sinus rhythm.

'N': other arrhythmia, including supraventricular tachycardia, sinus bradycardia, LBBB, RBBB, APC and PVC beats.

'Asyst': asystole; ECG signal with a peaktopeak amplitude of < 100 μV, lasting more than 4 s.

3.
Intermediate rhythms

'VTlo': slow ventricular tachycardia with a rate of < 180 bpm.

'fine VF': any VF signal with an amplitude in the range 100  200 μV.
It is clear from this classification that VT is divided into two categories according to heart rate; 'VThi' and 'VTlo'. This VT classification considers border heart rate as 180 bpm. It is, however, not strict. It may be in the range 150  180 bpm. AEDs only advise/deliver shock to shockable rhythms, and intermediate rhythms are treated in a different way called antitachycardia pacing.
Detection of VTVF from other arrhythmias
To detect the life threatening cardiac arrhythmias, VT and VF, from other arrhythmias, we propose to use a property that does not match with that of any nonVTVF signal. Typical ECG waveforms of NSR, VT and VF are given in Figure 1. Here, NSR is treated as the representative of nonVTVF signals. The three waveforms are plotted in the same scale. From this figure we see that the width of the QRS complex is different for different arrhythmias. For NSR, it is noticed that the QRS interval is normally 0.06  0.10 sec and in case of VT, the QRS complex is more wider (> 0.10 sec). In VF, no QRS complex is noticed. On the other hand, P waves are normal (upright and uniform) in the NSR waveform and in case of VT and VF signal, no P waves are observed [31].
The distinguishable morphological characteristics of these three groups, namely nonVTVF, VT and VF can be quantified using a term called the absolute strength of a signal. The absolute strength or the mean of absolute value (MAV) of a signal x(n) of length N is defined as
Here, n stands for the number of samples within the chosen length. In case of NSR, the main representative of the nonVTVF group, the duration of the QRS complex is small as compared to one ECG period as illustrated in Figure 1(a). It is also observed from this figure that the NSR signal level is low for most of the time in an ECG cycle. Therefore, the absolute signal level of the QRS complexes dominates in the summation of MAV calculation (eqn. (1)). A low MAV is thus obtained for such episodes. In case of VT, we see that the QRS complex is much wider than that of NSR, and the ECG signal hardly goes through the baseline as is the case for VF. Therefore, the MAV of VT and VF for a fixed duration window is comparatively larger than that for the NSR.
Before calculating the total MAV of a ECG signal, first it is necessary to normalize the ECG signal because the ECG signals collected from the different databases have different dynamic value. Another important thing to be noted is that, to use the MAV as the threshold parameter, we need to properly choose the analysis window duration. To understand the reason behind the necessity to appropriately choose the analysis window duration, consider a normalized VF episode of 8s length from cu 01m file of CU database shown in Figure 2. If we choose an 8s episode length, then it may include a damped VF signal, as shown in figure 2, where most of the signal samples fall in the low amplitude range, and the MAV becomes low (e.g., 0.2577). Therefore, it is necessary to make the analysis window length small. Choosing an analysis window of too small duration (say, 1s) creates the same problem as observed in the TCI method. Here, we choose the 2s window for analysis. After calculating the MAV of this 2s analysis window, we shift the analysis window by 1s successively for other segments of 2s within the 8s ECG episode and calculate the MAV again. After completion of shifting the analysis window to cover the whole decision frame, we average all the MAV s found in each stage and finally MAV = 0.34 is found which is higher than that obtained for the 8s analysis window. In this way, by appropriately selecting the analysis window length in calculating the MAV , we can overcome the effect of damped behavior of the ECG signal.
Observation of other nonVTVF ECG waveforms such as Premature Ventricular Contraction (PVC), Premature Atrial Contraction (PAC), Supraventricular Tachycardia (SVT) etc. reveals that these abnormalities also have low MAV compared to VT and VF. For example, PVC arrhythmia has small MAV because a PVC beat contains only wide QRS complex and no P waves or T waves are associated with this abnormal beat [31]. Thus, we can use MAV as the performance index to discriminate the VTVF from other arrhythmias.
In ECG analysis, it is important that we choose the episode length or decision frame appropriately. Decision frame should be taken in such a way that is neither too short to make a false alarm nor too long to cause severe cardiac arrest. Decreasing the episode length from its optimum value results in a low accuracy but quick detection. On the contrary, increasing the episode length improves the accuracy up to a certain level but requires longer detection time.
The whole process of separating VT plus VF from other arrhythmias can be described as in the following:

1.
Choose a segment of ECG signal of L _{ e } second duration. This segmented ECG signal of L _{ e } second duration should be stored for the second stage.

2.
The segment of the ECG signal is preprocessed using the wellknown filtering process as used in [32], which is carried out in a MATLAB routine, called filtering.m [33]. The filtering algorithm works in four successive steps.

First, the mean value is subtracted from the signal.

Second, a moving average filter is applied in order to remove the power line noise.

Third, a drift suppression is carried out by a high pass filter with a cutoff frequency of 1 Hz.

In the last step, a low pass Butterworth filter with a cutoff frequency of 30 Hz is applied in order to suppress the high frequency noise like interspersions and muscle noise.
All filters in the preprocessing step is implemented using the Matlab routine 'filtfilt' function.

3.
Then, choose a smaller segment x(n) from the ECG signal of L _{ e } second duration in such a way that the length of the segment is 2s. If the sampling frequency of the ECG signal is F _{ s } samples/s, then the total sample within this segment (N) is 2F _{ s } . For example, the sampling frequency of the ECG signal of the MITDB is 360 smaples/sec. Thus the length of the smaller segment N is 2 × 360 = 720 samples.

4.
Next, divide the smaller segment x(n) by the maximum absolute value found in that segment.

5.
Calculate the MAV using (1).

6.
Shift the window by 1s successively for other segments of 2s within the L _{ e } second ECG episode and go through step (4) to (5).

7.
Make decision on every L _{ e } second ECG episode (L _{ e } ≥ 2) by averaging the L _{ e }  1 consecutive values of MAV obtained from the L _{ e }  1 consecutive 2s segments with 1s step. The average value, MAV _{ a } for an L _{ e } second episode is calculated as
$$MA{V}_{a}=\frac{1}{{L}_{e}1}{\displaystyle \sum _{i=1}^{{L}_{e}1}M}A{V}_{i}$$(2)
where MAV _{ i } is the value of MAV in the ith 2s stage.
We calculate the MAV _{ a } of the three pathologies shown in Figure 1 and are obtained as 0.0765 (NSR), 0.3954 (VT) and 0.4116 (VF). To verify the effectiveness of the MAV index for separating the nonVTVF arrhythmias from the VTVF arrhythmias, other nonVTVF representatives namely, left bundle branch block beat, nodal (junctional) premature beat (rate ≈ 100 bpm), high rate supraventricular tachycardia (rate ≈ 100 bpm), premature ventricular contraction, right bundle branch block beat and paced beat are chosen from the ECG databases. These six pathologies are demonstrated in Figure 3 and their MAV _{ a } are 0.1649, 0.0954, 0.1372, 0.1475, 0.1571, 0.2166, respectively. Certainly, there is a clear separation of these MAV _{ a } values with those obtained from VT and VF episodes.
If MAV _{ a } is greater than a certain threshold MAV _{ d } , VTVF is detected. To determine the threshold value, training dataset is used. Figure 4 shows the probability distribution of MAV _{ a } of the training dataset and the test dataset. The threshold value is selected from the probability distributions of the training dataset shown in Figure 4(a) and we have chosen MAV _{ d } = 0.27 for L _{ e } = 8s to ensure high specificity and also good sensitivity. It is also noticed from Figure 4(b) that when we apply this threshold to the test dataset, high accuracy is still obtained.
Separation of VF from VTVF
Now that we have separated VTVF from other arrhythmias. In this stage, we separate VF from VT. Before we explain our motivation for using the EMD technique, we briefly describe what EMD is.
EMD Preliminaries
EMD is a signal decomposing method which is fully datadriven and does not require any a priori basis function [27, 34]. The aim of the EMD is to decompose the signal into a sum of intrinsic mode functions (IMFs). An IMF is a function that satisfies two conditions: (1) in the whole data set, the number of extrema and the number of zero crossings must either be equal or differ at most by one; and (2) at any point, the mean value of the envelop defined by the local maxima and the envelop defined by the local minima is zero. An IMF represents the oscillatory mode embedded in the data as a counterpart to the simple harmonic function used in Fourier analysis [35].
Given a signal x(n), the starting point of the EMD is the identification of all the local maxima and minima. All the local maxima are then connected by a cubic spline [36] curve as the upper envelop e _{ u } (n). Similarly, all the local minima are connected by a spline curve as the lower envelop e _{ l } (n). The mean of the two envelops is denoted as m _{1}(n) = [e _{ u } (n)+e _{ l } (n)]/2 and is subtracted from the signal. Thus the first component h _{1}(n) is obtained as
The above procedure to extract the IMF is called the sifting process. Ideally, h _{1}(n) should be an IMF, as the construction of h _{1}(n) seems to have been made to satisfy all the requirements of IMF. Since h _{1}(n) still contains multiple extrema in between zero crossings, the sifting process is performed again on h _{1}(n). This process is applied repetitively to the proto IMF h _{ k } (n) until the first IMF c _{1}(n), which satisfies the IMF condition, is obtained. Couple of stopping criteria are used to terminate the sifting process [27]. A commonly used criterion is the value of standard deviation, SD, computed from the two consecutive sifting:
where, N is the total number of samples in x(n). When the SD is smaller than a threshold, the first IMF c _{1}(n) is obtained. Then c _{1}(n) is separated from the rest of the data by
It is to be noted that the residue r _{1}(n) still contains some useful information. We can therefore treat the residue as a new signal and apply the same sifting process to obtain
The whole procedure terminates when either the component c _{ q } (n) or the residue r _{ q } (n) becomes very small or when the residue r _{ q } (n) becomes a monotonic function. Combining (5) and (6) yields the EMD of the original signal,
The results of the decomposition are q  intrinsic modes and a residue. The lower order IMFs capture the fast oscillation modes while the higher order IMFs typically represent the slow oscillation modes present in the underlying signal [27, 37]. An example illustrating the Empirical Mode Decomposition is given in the 'Appendix' section.
As mentioned earlier, the VT waveform contains the QRS complex but it is absent in the VF waveform. The asymmetry of the QRS complex with respect to the baseline gives rise to asymmetric signal envelopes which are comprised of local maxima and minima. Another interesting thing to be noted is that in case of VT, comparatively short duration of the QRS complex results in a wideband ECG signal. On the other hand, the QRS complex is absent in VF and as a result this pathology has more symmetric envelopes than do other abnormalities and thus possesses narrowband characteristics. Therefore, to separate VF from VT, the EMD technique can effectively use the factors of narrowband/wideband characteristics and symmetry/asymmetry property of a signal's envelopes. Now, we apply the EMD technique on a VF episode to decompose it into IMFs and plot the original ECG signal x(n) along with its first IMF as shown in Figure 5(a). From Figure 5(a) we can say that in case of VF, its first IMF is very much close to the original ECG signal. This is because the VF has certain properties that well match the properties of the IMF as stated above. As the EMD technique cannot decompose an IMF signal further, therefore, in case of a VF episode, there is a unique relationship between the ECG signal and its first IMF. Here, unique relationship means that the original ECG signal and its first IMF is very much similar. In some cases high frequency noise still remains in the ECG signal after preprocessing. Therefore, when we apply EMD to decompose the VF signal, the first IMF captures this high frequency noise as the fast oscillation mode illustrated in Figure 5(b). To overcome this effect we consider the sum of the first two IMFs instead of using only the first one. We can observe from Figure 5(b) that unique relationship still exists between the ECG signal and the sum of first two IMFs for the VF episode. In case of VT, this unique relationship or similarity between the ECG signal and the sum of its first two IMFs does not hold as illustrated in Figure 6 for both noise free and noise corrupted VT signals.
To exploit the property of unique relationship between the ECG signal and the sum of its first two IMFs that exists in case of the VF only, sum of the first two IMFs from the ECG signal is subtracted and the MAV of the difference signal is calculated. Since, the dynamic range of the ECG signal varies from database to database, we normalize this MAV with respect to the original ECG signal. In case of a VF episode, the normalized MAV or NMAV of the difference signal is very small than that of a VT episode. Here, we choose a 2s analysis window as in the previous case. But in this case, the performance index (NMAV ) is less sensitive to the analysis window length.
The process of detecting VF from VTVF can then be described as below:

1.
First, choose a segment x(n) of duration 2s and N samples from the previously saved ECG signal of L _{ e } second duration.

2.
At this stage, the ECG signal is preprocessed in three successive steps.

First, the mean value is subtracted from the signal.

Second, a drift suppression is carried out by a highpass filter with a cutoff frequency of 1 Hz.

In the last step, a lowpass Butterworth filter with a cutoff frequency of 20 Hz and order 12 is applied to suppress the high frequency information.

3.
Apply EMD on x(n) and determine
$$im{f}_{12}(n)=im{f}_{1}(n)+im{f}_{2}(n)$$
where, imf _{1}(n) and imf _{2}(n) denotes the first and second IMFs, respectively.

4.
Then, calculate the difference between the original signal and sum of its first two IMFs,
$$e(n)=x(n)im{f}_{12}(n)$$ 
5.
The normalized MAV of e(n) used as the index for discriminating VF from VT is calculated as
$$NMAV=\frac{{\scriptscriptstyle \frac{1}{N}}{\displaystyle \sum _{n=0}^{N1}\lefte\left(n\right)\right}}{{\scriptscriptstyle \frac{1}{N}}{\displaystyle \sum _{n=0}^{N=1}\leftx\left(n\right)\right}}$$ 
6.
Shift the window by 1s successively for other segments of 2s within the L _{ e } second ECG episode and go through step (ii) to (iv).

7.
Make decision on every L _{ e } second ECG episode (L _{ e } ≥ 2) by averaging L _{ e }  1 consecutive values of NMAV obtained from L _{ e }  1 consecutive 2s data segments with 1s step. The average value NMAV _{ a } for an L _{ e } second episode is calculated as
$$NMA{V}_{a}=\frac{1}{{L}_{e}1}{\displaystyle \sum _{i=1}^{{L}_{e}1}N}MA{V}_{i}$$(8)
where NMAV _{ i } is the value of NMAV in the ith 2s stage.
Applying the above stated process, the NMAV _{ a } are obtained as 0.08 (for Figure 5(b)), 0.97 (for Figure 6(a)) and 0.93 (for Figure 6(b)). If NMAV _{ a } is less than a certain threshold NMAV _{ d } , VF is detected, otherwise VT is detected. The threshold value NMAV _{ d } is selected by a process as described before using the training dataset. As in this stage we separate VF from VT, therefore, the training and test datasets include only VF and VT episodes. This threshold value is then applied to the test dataset. Figure 7 shows the probability distribution of NMAV _{ a } of the training dataset and the test dataset. From the training dataset, we have chosen NMAV _{ d } = 0.65 for L _{ e } = 8s to ensure that both VF and VT detection accuracies are good. It is also noticed from Figure 7(b) that the threshold value calculated from the training dataset can be applied to the test dataset maintaining almost the same accuracy as found from the training dataset.
Classification of VT and VF according to the AHA recommendations
As only the certain classes of VTs and VFs require highenergy shock for treatment, it is necessary to classify the VT and VF according to the heart rate and amplitude, respectively. Since, the heart rate calculation is complicated than the amplitude determination, hence at first we propose a technique to determine the heart rate. The heart rate in bpm is defined as the number of QRS complexes that occur in 60 sec. To determine the heart rate of an ECG signal, first derivative of the ECG signal is utilized. The reason behind the choice of the first derivative of the ECG signal is to utilize the high slope of the QRS complex. Figs. 8(a) and 8(b) show the VT signal and its first derivative. Figure 8(b) illustrates that when QRS complexes occur, correspondingly there is a high value (both in positive and negative part) in the first derivative signal. We consider only the positive part of the first derivative signal. Then this signal is filtered to enhance the QRS complexes further. From this filtered signal shown in Figure 8(c), the heart rate is easily calculated. The whole process of determining the heart rate of the ECG signal is described below:

1.
First, choose a segment x(n) of duration L _{ e } second and N samples from the previously saved ECG signal and then perform preprocessing as stated in Section.

2.
Calculate the first derivative (x _{ d } (n)) of x(n).
$${x}_{d}(n)=x(n)x(n1)$$
The waveform of x _{ d } (n) is shown in Figure 8(b).

3.
Keep only the positive part of x _{ d } (n).
$${x}_{dp}(n)=\{\begin{array}{cc}\begin{array}{r}\hfill {x}_{d}(n)\\ \hfill 0\end{array}& \begin{array}{l}if{x}_{d}(n)\ge 0\hfill \\ ;otherwise\hfill \end{array}\end{array}$$ 
4.
Apply the moving average filter on x _{ dp } (n) and find x _{ dpf } (n).
$${x}_{dpf}(n)={\displaystyle \sum _{k=0}^{k=\alpha}x}(nk)$$
where, α = F _{ s } /10 and F _{ s } is the sampling frequency. If α is not an integer, then it is rounded to the nearest integer value. The waveform of x _{ dpf } (n) is shown in Figure 8(c).

5.
Determine the maximum value (C) and the corresponding peak index (I) of x _{ dpf } (n) and calculate the threshold value (T _{ h } ) from C.
$$\begin{array}{lll}C\hfill & =\hfill & \mathrm{max}\{{x}_{dpf}(n)\}\hfill \\ {T}_{h}\hfill & =\hfill & C\times \beta \hfill \end{array}$$
where, β is a properly chosen constant. Here, we choose β = 0.25.

6.
Store the peak index (I) and mask x _{ dpf } (n) around this position.
$${x}_{dpf}(I\gamma :I+\gamma )=0$$
where, γ = F _{ s } /8; if γ is not an integer, then it is rounded to the nearest integer value.

7.
Now, calculate again the maximum value (C) of x _{ dpf } (n) and go through step (vi) until C goes below the T _{ h } .

8.
Determine the total number of peaks (N _{ p } ) those are above T _{ h } and calculate H _{ R } .
$${H}_{R}=\frac{{N}_{p}\times 60}{{L}_{e}}bpm$$
If the heart rate of the VT signal is greater than 180 bpm, then this VT is called the shockable VT. As the decision of shockable or intermediate VT is dependent on the heart rate of the episode, hence, we calculate the total number of QRS beats in a episode. Now, to check the efficiency of the heart rate determination algorithm, two episodes selected are shown in Figs. 8(e)(f). At first, the total number of QRS beats in these episodes are determined from the annotation. Then, the proposed derivative based heart rate determination algorithm is used to calculate the total number of QRS beats and it is found to be 15 beats for Figure 8(e) and 17 beats for Figure 8(f). In both cases, the total number of QRS complexes obtained by using our algorithm are the same as determined from the annotation. Thus, this heart rate determination method, though simple, may be used to calculate the heart rate of an ECG episode. However, in more complicated cases any standard heart rate determination algorithm reported in the literature [38, 39] may be adopted to classify the VT. On the other hand, the amplitude of the VF signal is determined by taking the maximum value of the absolute VF signal within a episode. If the amplitude is greater than 200 μV, than this VF is called the coarse VF.
Quality Parameters
The quality parameters, we have used for the assessment of algorithms, are sensitivity, speci city, positive predictivity, and accuracy. For 'VTVF' detection, the first four parameters are defined by
For 'VF' and 'shockable rhythm' detection, the definition of these four quality parameters contain 'VF' and 'shockable rhythm' in place of 'VTVF', respectively. While calculating these four quality parameters to judge the effectiveness of an algorithm, in case of any unsatisfactory results obtained, the values of the respective thresholds were adjusted in order to obtain the best possible results.
Results and Discussion
The full classification of different ECG pathologies is shown in Figure 9. To compare our algorithm with other reported algorithms in the literature, our classification approach can be interpreted as three different ECG arrhythmias identification schemes; such as

1.
VTVF and nonVTVF

2.
VF and nonVF (nonVTVF+VT)

3.
shockable (VF+VT above 180 bpm) and nonshockable (nonVTVF+VT below 180 bpm)
Since the annotated files do not contain enough lowamplitude signals (fine VF), therefore, this type of signal is not addressed in this identification scheme and the VF signals in the shockable rhythms are actually coarse VF. Now, this section is divided into three subsections and each subsection presents the results of each identification scheme.
Detection of VTVF from other arrhythmias
First, we test the separability of our algorithm between the two classes of ECG signals, i.e., 'VTVF' and 'nonVTVF' against the annotated decisions suggested by the cardiologists in the respective databases. We compare our algorithm with the complexity measure algorithm [10] and the results are shown in Table 1. Comparative results illustrate that our algorithm shows better performance than the complexity measure algorithm. Also notice that the accuracy of the proposed MAV scheme is significantly higher than that of the complexity measure algorithm. Thus our simple and fast algorithm can separate VTVF from nonVTVF with higher specificity and sensitivity simultaneously. In this case, we had to change the threshold value of the CPLX algorithm from that defined in [10] to obtain higher sensitivity.
Detection of VF from other arrhythmias
VF occurs at the clinically crucial stage of human being. As mentioned earlier, while detecting VF from the other arrhythmias in the first stage, we should make the specificity high because a low specificity may risk patient's life by generating a false alarm to provide a high energy shock as treatment to save his/her life. But our proposed sequential algorithm leads to a high specificity. Quality parameters of our proposed algorithm with some other wellknown algorithms are shown in Table 2. It is clear that our algorithm shows higher sensitivity compared to all other algorithms with very good specificity (99.32%). To obtain higher specificity, we had to change the critical threshold parameter of the HILB and PSR methods from that defined in the respective papers. To compare different methods independent of the value of the decision thresholds, the critical threshold parameter in the decision stage of the algorithm is varied. By varying the threshold, we can vary specificity and sensitivity as shown in Table 3. This table illustrates that our proposed method performs much better than the other VF detection algorithms.
Detection of shockable rhythms from other arrhythmias
This subsection presents the results of our last identification scheme which classifies the ECG pathologies into two groups: shockable and nonshockable rhythms. To compare our algorithm with the reported algorithm in [5], some modifications in the threshold values are made to accommodate unequal episode lengths (L _{ e } ). Our proposed algorithm considers L _{ e } = 8s where L _{ e } = 10s was considered in [5]. Modifications are shown in Table 4. For example, if Count 1 < 250 for L _{ e } = 10s, then for L _{ e } = 8s Count 1 < 250 10 * 8 or Count 1 < 200.
Here, as we concentrate only on the shockable and nonshockable rhythms, some classification errors may not result in detection errors. For example, from Figure 10 we see that a classification error occurs when a VT above 180 bpm is falsely detected as VF in the second stage. Since a VT above 180 bpm (one type of shockable rhythms) is falsely mapped into the VF group, which is also in the class of shockable rhythms, therefore, this classification error does not make any detection error as long as shockable rhythm is our concern. Using the modified threshold values mentioned in Table 4, the results obtained are presented in Table 5. As can be seen, our algorithm performs better than the count [5] algorithm in every index in detecting the shockable rhythms correctly.
Conclusions
A novel method for the identification of life threatening cardiac abnormalities from other arrhythmias has been presented. Performing sequential signal processing, we have detected these cardiac abnormalities with good accuracy. It has been shown that the proposed algorithm based on the MAV parameter and EMD technique can detect the VT plus VF signals correctly from other arrhythmias, and the accuracy level remains higher than that of other reported techniques. The effectiveness of the proposed technique has been demonstrated using standard databases over a vast range of both normal and abnormal ECG records. The MAV index successfully separates the VTVF arrhythmias from different types of abnormalities. And the other parameter NMAV which is calculated using the IMFs of the EMD technique can successfully separate VF from VTVF. Finally, a fast and simple heart rate determination technique is used to separate the high rate VT. Consistent results have been obtained by applying our algorithm on different wellknown databases namely, MITBIH database, CU database and MITBIH Malignant Ventricular Arrhythmia database. Determination of the threshold parameters from the training dataset and then their successful application on the test dataset proves that the proposed parameters are universal. Some signal episodes were very difficult for classification even by expert cardiologists. Accuracy of our proposed technique slightly falls due to these confusing episodes. The algorithm presented here has strong potential to be applied in clinical applications for accurate detection of life threatening cardiac arrhythmias.
Appendix
The steps involved in the EMD technique are described below using an example.

1.
Determine the upper envelop e _{ u } (n) and the lower envelop e _{ l } (n). These two envelopes are shown in Figure 11(a) along with the original signal x(n).

2.
Determine the mean of the envelope, i.e., m _{1}(n) = [e _{ u } (n) + e _{ l } (n)]/2. The variation of m _{1}(n) is displayed in Figure 11(a).

3.
Extract the first component h _{1}(n) using eqn. (3).

4.
Ideally, h _{1}(n) should be the first IMF. But, it is observed from Figure 11(b) that the h _{1}(n) does not satisfy the conditions of an IMF.

5.
Now, treat h _{1}(n) as x(n) in step (1). Determine the two envelopes from h _{1}(n) and the mean of these two envelopes (Figure 11(b)). After subtraction of the mean from the h _{1}(n) a new signal h _{1(2)}(n) is obtained. Now, check the conditions of an IMF and also calculate the value of SD from eqn. (4), where h _{1(1)}=h _{1}(n).

6.
Continue the process until h _{1(k1)}satisfies the conditions of the IMFs. When the conditions are satisfied, the first IMF is found as shown in Figure 11(c). Now, the first IMF is subtracted from the initial signal x(n).

7.
The second IMF (Figure 11(d)) is extracted following the steps (1) to (6) except that the subtracted signal is used instead of x(n). No further decomposition is performed here as we need two IMFs for our analysis. The residue of the EMD is shown in Figure 11(e).
References
 1.
Amann A, Tratnig R, Unterkofler K: A new ventricular fibrillation detection algorithm for automated external defibrillators. Comput Cardiol 2005, 32: 559–562. full_text
 2.
Amann A, Tratnig R, Unterkofler K: Detecting ventricular fibrillation by timedelay methods. IEEE Trans Biomed Eng 2007, 54: 174–177. 10.1109/TBME.2006.880909
 3.
Barro S, Ruiz R, Cabello D, Mira J: Algorithmic sequential decisionmaking in the frequency domain for life threatening ventricular arrhythmias and imitative artifacts: a diagnostic system. J Biomed Eng 1989, 11: 320–328. 10.1016/01415425(89)900678
 4.
Thakor NV, Zhu YS, Pan KY: Ventricular tachycardia and fibrillation detection by a sequential hypothesis testing algorithm. IEEE Trans Biomed Eng 1990, 37(9):837–843. 10.1109/10.58594
 5.
Jekova I, Krasteva V: Real Time detection of ventricular fibrillation and tachycardia. Physiol Meas 2004, 25: 1167–1178. 10.1088/09673334/25/5/007
 6.
Jekova I, Krasteva V, Ménétré S, Stoyanov T, Christov I, Fleischhackl R, Schmid JJ, Didon JP: Bench study of the accuracy of a commercial AED arrhythmia analysis algorithm in the presence of electromagnetic interferences. Physiol Meas 2009, 30: 695–705. 10.1088/09673334/30/7/012
 7.
Gauna SRd, Lazkano A, Ruiz J, Aramendi E: Discrimination between ventricular tachycardia and ventricular fibrillation using the continuous wavelet transform. Computers in Cardiology 2004., 31:
 8.
Caswell S, Thompson J, Jenkins J, DiCarlo L: Separation of ventricular tachycardia from ventricular fibrillation using paired unipolar electrograms. Cornputen in Cardiology 1996.
 9.
Arafat MA, Sieed J, Hasan MK: Detection of ventricular fibrillation using empirical mode decomposition and Bayes decision theory. Computers in Biology and Medicine 2009, 39(11):1051–1057. 10.1016/j.compbiomed.2009.08.007
 10.
Zhang XS, Zhu YS, Thakor NV: Detecting ventricular tachycardia and fibrillation by complexity measure. IEEE Trans Biomed Eng 1999., 46(5):
 11.
Jekova I, Mitev P: Detection of ventricular fibrillation and tachycardia from the surface ECG by a set of parameters acquired from four methods. Physiol Meas 2001, 23: 629–634. 10.1088/09673334/23/4/303
 12.
Didon J, Dotsinsky I, Jekova I, Krasteva V: Detection of Shockable and NonShockable Rhythms in Presence of CPR Artifacts by TimeFrequency ECG Analysis. Computers in Cardiology 2009, 36: 817–820.
 13.
Chen S, Thakor NV, Mover MM: Ventricular fibrillation detection by a regression test on the autocorrelation function. Med Biol Eng Comput 1987., 25: 10.1007/BF02447420
 14.
Langer A, Heilman MS, Mower MM, Mirowski M: Considerations in the development of the automatic implantable defibrillator. Med Instrum 1976., 10:
 15.
Kuo S, Dillman R: Computer detection of ventricular fibrillation. Comput Cardiol 1978.
 16.
RH Clayton AM, Campbell RWF: Comparison of four techniques for recognition of ventricular fibrillation from the surface ECG. Med Biol Eng Comput 1993., 31:
 17.
Heij S, Zeelenberg C: A fast realtime algorithm for the detection of ventricular fibrillation. Comput Cardiol 1987, 707–710.
 18.
Jenkins J, Noh KH, Guezennec A, Bump T, Arzbaecher R: Diagnosis of atrial fibrillation using electrograms from chronic leads: Evaluation of computer algorithms. PACE 1988, 11: 622–631.
 19.
Ripley KL, Bump TE, Arzbaecher RC: Evaluation of techniques for recognition of ventricular arrhythmias by implanted devices. IEEE Trans Biomed Eng 1989, 36: 618–624. 10.1109/10.29456
 20.
Thakor NV, Natarajan A, Tomaselli G: Multiway sequential hypothesis testing for tachyarrhythmia discrimination. IEEE Trans Biomed Eng 1994, 41: 480–487. 10.1109/10.293223
 21.
Chen SW, Clarkson PM, Fan Q: A robust sequential detection algorithm for cardiac arrhythmia classification. IEEE Trans Biomed Eng 1996, 43: 1120–1125. 10.1109/10.541254
 22.
Lin D, DiCarlo LA, Jenkins JM: Identification of ventricular tachycardia using intracavity ventricular electrograms: Analysis of time and frequency domain patterns. PACE 1988, 11: 1592–1606.
 23.
Throne RD, Jenkins JM, Winston SA, DiCarlo LA: A comparison of four new timedomain techniques for discriminating monomophic ventricular tachycardia form sinus rhythm using ventricular waveform morphology. IEEE Trans Biomed Eng 1991, 38: 561–570. 10.1109/10.81581
 24.
Arafat MA, Chowdhury AW, Hasan MK: A simple time domain algorithm for the detection of ventricular fibrillation in electrocardiogram. Signal, Image and Video Processing 2009.
 25.
Werther P, Klotz A, Granegger M, Baubin M, Feichtinger HG, Amann A, Gilly H: Strong corruption of electrocardiograms caused by cardiopulmonary resuscitation reduces efficiency of twochannel methods for removing motion artefacts in nonshockable rhythms. Resuscitation 2009, 80(11):1301–1307. 10.1016/j.resuscitation.2009.07.020
 26.
Jekova I: Comparison of five algorithms for the detection of ventricular fibrillation from the surface ECG. Physiol Meas 2000, 21: 429–439. 10.1088/09673334/21/4/301
 27.
Huang NE, Shen Z, Long SR, Wu MC, Shih HH, Zheng Q, Yen NC, Tung CC, Liu HH: The empirical mode decomposition and hilbert spectrum for nonlinear and nonstationary time series analysis. Proc R Soc Lond 1998., 454:
 28.
Massachusetts Institute of Technology, MITBIH arrhythmia database [http://www.physionet.org/physiobank/database/mitdb]
 29.
Massachusetts Institute of Technology, CU database [http://www.physionet.org/physiobank/database/cudb]
 30.
Massachusetts Institute of Technology, MITBIH Malignant Ventricular Fibrillation database [http://www.physionet.org/physiobank/database/vfdb]
 31.
Jones SA: ECG Notes. Philadelphia: Nursing: Lisa Deitch; 2005.
 32.
Amann A, Tratnig R, Unterkofler K: Reliability of old and new ventricular fibrillation detection algorithms for automated external defibrillators. Biomed Eng Online 2005., 4(60): [http://0www.biomedcentral.com.brum.beds.ac.uk/content/pdf/1475–925x4–60.pdf]
 33.
filtering.m [https://homepages.fhv.at/ku/karl/VF/filtering.m]
 34.
Junsheng C, Dejie Y, Yu Y: Research on the intrinsic mode function (IMF) criterion in EMD method. Mechanical Systems and Signal Processing 2006, 20(4):817–824. 10.1016/j.ymssp.2005.09.011
 35.
Flandrin P: TimeFrequency/TimeScale Analysis. Academic Press; 1999.
 36.
Gregory JA: Shape preserving spline interpolation. ComputerAided Design 1986, 18: 53–57. 10.1016/S00104485(86)800124
 37.
Sharpley RC, Vatchev V: Analysis of the Intrinsic Mode Functions. Constructive Approximation 2006, 24: 17–47. 10.1007/s003650050603z
 38.
Afonso VX, Tompkins WJ: ECG Beat Detection Using Filter Banks. IEEE Transactions on Biomedical Engineering 1999, 46(12):192–202. 10.1109/10.740882
 39.
Arafat MA, Hasan MK: Automatic Detection of ECGWave Boundaries using Empirical Mode Decomposition. Proc ICASSP 2009, 461–464.
Acknowledgements
This work was supported in part by the National Research Foundation (NRF) of Korea funded by the Korean government (MEST) (No: 20090078310).
Author information
Affiliations
Corresponding author
Additional information
Competing interests
The authors declare that they have no competing interests.
Authors' contributions
EM carried out the implementation of the idea, contributed in the development of new characteristic index for discriminating different pathology of signals, collected data from different standard databases for analysis, and drafted the manuscript. SY participated in the design of the study and interpretation of data, and was involved in critically revising the manuscript. MK conceived of the study, and participated in its design, analysis and interpretation of data, defined mathematical index to be used for discriminating arrhythmias, and helped to draft and finalize the manuscript. All authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
About this article
Cite this article
Anas, E.M.A., Lee, S.Y. & Hasan, M.K. Sequential algorithm for life threatening cardiac pathologies detection based on mean signal strength and EMD functions. BioMed Eng OnLine 9, 43 (2010). https://0doiorg.brum.beds.ac.uk/10.1186/1475925X943
Received:
Accepted:
Published:
DOI: https://0doiorg.brum.beds.ac.uk/10.1186/1475925X943
Keywords
 Ventricular Tachycardia
 Ventricular Fibrillation
 Empirical Mode Decomposition
 Normal Sinus Rhythm
 Shockable Rhythm